# Land Surveyors United

Earth's Largest Land Surveyor Community

# WHICH METHOD OR INSTRUMENT ARE MOST ACCURATE USING GPS OR TOTAL STATION?

I'm using trimble R8 gps for subdividing of 500 plot with different area per plot in an open and flat area of around 3 square kilometer, but when i tried to use my total station (Trimble 5600 DR200) with same Benchmarks there are a shifting of around 10 centimeter,area of the plot is same as per coordinates computation methods,only the shifting is the main problem.(find the attachment)..

please to all well experience land surveyor and government surveyor give me some advice.thanks and more power to the LSU.gps%20and%20total%20station.pdfgps%20and%20total%20station%202.pdf

and

This Content Originally Published by a land surveyor to Land Surveyors United Network

Views: 2776

### Replies to This Discussion

I'll pass this along and try to share it with surveyors that I know will have the answer, but in the future, you might consider posting this in the Trimble Groupin order to get targeted feedback from Trimble Users.

I think that in both methods you'll get less than 10cm, in your case it seems to be a different set of coordinates in the benchmarks.

If you are using RTK sometimes the correction in the vectors tend to move the rover in the "same direction" over a period of time, you should take some control points to check that.

Also check the 3D distances (point to point) with a total station and watch if they are the same as the GPS.

In my point of view, It should not have such discrepancy. I believe that the survey procedure have a problem. If i will work for that, i should establish a control network first using GPS and do proper adjustment. Afterwards, do setting-out using total station or rtk using adjusted control points.

the bench mark that i am using is established by the government with high precision of x,y and z.

setting out using adjusted coordinates are ok but we cannot say which are the good method between gps or total station.but anyway thanks for the reply, i will focus now using rtk survey with R8 gps.

Mr Eric i am using benchmark established by government with high precision of x,y and z, R8 base and total station are setting that benchmark but still 10cm discrepancy,and also that 2 instruments are new calibrated,but anyway thanks for the reply,now i will focus using R8 gps with rtk survey maybe gps is more accurate than total station.

Julius, it is a basic for several contract that the client provides primary control points that will be initially used for all survey works. Therefore, for such big area, you should do your circuit and close it based on the provided references, with the client or engineer's approval, thus, utilizing the new control points for setting-out.

Your instruments are of good specs, meaning, should have good survey result. The one i mentioned earlier is a basic construction procedure. Anyway, if you want to utilize only a single baseline (Pts. provided by the client), and considering the type of work, better to use RTK, otherwise, using a total station without applying corrections and no adjustment will cause accumulation of errors resulting to erroneous survey.

For me, it should not what instrument is better that was put in question, it is on the survey procedure.

I hope you got my point... Good luck...

ok sir thank you very much i got your point.

this is a scale factor problem .. we need to calculate scale factor and we need to enter the value inside the total station

In my opinion, the difference is a simple geodesy issue. RTK positioning is a truely geodetic work while total station positioning is a locally based endeavor. For that reason, I would say that there are seven basic transformation factors that caused the difference.

Where Thousands of Professional Land Surveyors, Students of Surveying and Educators are United through Collaborative Knowledge and Purpose.

## Land Surveyors Forum

### Survey Abstract Coordinates

Started by John J Rodriquez in Introduce Yourself to the Community. Last reply by Kevin D. Allen 14 hours ago.

### Control surveying problem

Started by shree ram bhandari in Land Surveying Jobs. Last reply by Billy Brooks 3 hours ago.

### Trimble Installation Manager help

Started by Jeff Johnstone in Surveying Equipment and Software Talk. Last reply by Jaybird on Monday.

### Magnetic North Update

Started by Billy Brooks in Surveying Best Practices. Last reply by Billy Brooks Feb 6.

### AERIAL STEREOSCOPY

Started by Lalith Senanayake in Surveying Related Links. Last reply by shree ram bhandari on Tuesday.

### Standard Symbols for site facilities

Started by Mohammad Salem in Surveying Equipment and Software Talk. Last reply by ⚡Survenator⌁ Jan 28.

Started by Ian harvey in Surveying Equipment and Software Talk. Last reply by Ian harvey Jan 31.

### Study Guide - Reference Books

Started by Ron Willhoit in Introduce Yourself to the Community. Last reply by ⚡Survenator⌁ Jan 27.

### Job Quotation

Started by Nitesh Shrestha in Land Surveying Jobs. Last reply by Nitesh Shrestha Jan 21.

### Survey under tree canopy

Started by Massimo in Surveying Best Practices. Last reply by Maurice Heter 2 hours ago.

### Mean Sea Level (MSL)

Started by Chuck Turlington in Geospatial Industry Types. Last reply by Kim Noblitt Jan 17.

### Can I convert TPS1200 codes to TPS1100

Started by Maciej Michalak in Surveying Equipment and Software Talk. Last reply by ⚡Survenator⌁ Jan 8.

### VIP Codes for TPS1100

Started by Maciej Michalak in Introduce Yourself to the Community. Last reply by ⚡Survenator⌁ Jan 8.

### GPS Wobbly Breaklines - has anybody encountered this?

Started by Carl Strydom in Surveying Field Tips and Tricks. Last reply by WADE DEDOBBELEER Jan 21.

### Surface pro

Started by Malik in Surveying Field Tips and Tricks. Last reply by Kevin D. Allen Jan 7.

### Leica TPS1100 Code list Problem

Started by Maciej Michalak in Surveying Equipment and Software Talk. Last reply by ⚡Survenator⌁ Jan 9.

### Seeking for a job in Brisbane Australia

Started by Najib Lishaa in Introduce Yourself to the Community. Last reply by Jaybird Jan 4.

### ROOF PEAK SURVEY

Started by Mthulisi Mloyi in Professional Advice for New Surveyors. Last reply by ⚡Survenator⌁ Jan 4.

### As- built pipeline surveying

Started by Alyssa Fournier in Surveying Field Tips and Tricks. Last reply by Alyssa Fournier Jan 4.

### laptop in the field

Started by Malik in Introduce Yourself to the Community. Last reply by Lyubo David Jan 21.

Tweet to us

## Tools,Apps and Quick Guides

DEDICATED TO THE MEMORY OF SKIP FARROW (1953-2015)

Help With Survey Equipment

#### Social Support For Equipment

Surveying Video Categories

Surveying Photo Categories

United States Hub Forums

 AK AL AR AZ CA CO CT DE FL GA HI IA ID IL IN KS KY LA MA MD ME MI MN MO MS MT NC ND NE NH NJ NM NV NY OH OK OR PA RI SC SD TN TX UT VA VT WA WI WV WY

International Surveyor Hub Forums

Resources for Company Owners

Land Surveyor Social Media

Surveyor Resources Collections

71 members

70 members

14 members

29 members

41 members