Earth's Largest Land Surveyor Community
Hy guys,i have a problem with my total station Leica TCR 1205+
for example,im doing stakeout points,after one hour work i get some problems.When i want to change a point which a want ti stakeout the time for a change point i so long ( maybe 30sec),i get same problem when i want to check point by ''dist'' taster.I have to wait more than 20sec to get a dist from point
I changed a memory card (i was trying with another one)
I reset RAM memory from total staion ( i did it in total station fuction,i didnt connect with PC)
This Content Originally Published by a land surveyor to Land Surveyors United Network
Usually a file dump will fix this problem. Particularly if it is a large stakeout job and you have stored a large number of points. Decide which points you can archive, and save them to a computer. Once saved delete them out of the job on your controller.
what is a dump file?
I had similar issues before. I had a pretty beat up prism and bought a new one. Worth a shot if you have an extra one laying around.
thank you, colleagues
i have only 50-6- points in job.Also im using Leica prisms
First,i will try to fix a dump file. if i will get same problem a have to check a hardware
what do you think about hardware? Maybe is main-plate?